# Engineering Resources

- Definition of Terms
- White Papers
- Pressure Point #1 Pressure Measurement Types
- Pressure Point #2 Understanding Accuracy and Precision
- Pressure Point #3 Linearity Measurements for MEMS Pressure Sensors
- Pressure Point #4 Dual Die Compensation for MEMS Pressure Sensors
- Pressure Point #5 Special Considerations for Mounting and Handling Pressure Sensors
- Pressure Point #6: Position Sensitivity in Pressure Sensors
- Pressure Point #7: Understanding Common-Mode Differential Pressure
- Pressure Point #8: Bandwidth vs. Signal to Noise Tradeoff
- Pressure Point #9: Pressure Sensor Technologies
- Pressure Point 10: Media Capability
- Warm Up Drift
- Design Considerations

- Technical Discussions
- Sensor History
- Product Catalogs PDF
- Technical Assistance
- Join Our Newsletter
- Sample Request

## Pressure Measurement Types

All Sensors ** Pressure Points** are application tips to simplify designing with microelectromechanical (MEMS) pressure sensors and avoiding common pitfalls.

*Pressure Point 1:***MEMS Pressure Sensors - Pressure Measurement Types**

Microelectromechanical system (MEMS) pressure sensors have changed the way that system designers and application engineers measure pressure. The simplicity of use, small size, low cost and ruggedness allow these sensors to address applications in automobiles and industrial process control as well as medical and handheld portable products. For example, a highly accurate altitude measurement in a handheld navigation device such as a smartphone that has three-axis accelerometers, gyroscope and magnetometers adds a tenth degree of freedom. Pressure sensing allows the navigation device to locate the exact floor of a destination.

- MEMS pressure sensors typically measure the pressure difference across a silicon diaphragm. As shown in Figure 1, there are three types of measurements:
- Gauge or gage (a) , where the pressure is referenced to the atmosphere
- Absolute (b) where the pressure is measured against a reference vacuum sealed inside the chip assembly
- Differential (c), where the pressure is measured as the difference between two pressures (delta P or ΔP)

**Figure 1.** Gage, absolute and differential pressure measurements have different types of pressure on each side of the diaphragm.

In these designs, the diaphragm is etched from silicon using micromachining, a chemical etching process. The measurement technique can involve capacitive or resistive (piezoresistive or pressure sensitive resistors). Piezoresistive designs are shown in Figure 1. Vacuum is a negative gauge pressure or a value below atmospheric pressure. When specifying or discussing the type of pressure measurement, it is essential to identify the type of measurement to convey an accurate description of the measurement technique. Table 1 shows the sensor requirements for several common measurements.

Application requirements for absolute, gauge and differential pressure sensing

**Table 1.** Common pressure measurement versus type of measurement.

**Atmospheric Pressure and Altitude**

Perhaps the most fundamental pressure measurement is atmospheric pressure. Standard atmospheric pressure at sea level is 29.92 inches of mercury (Hg) (760 mm Hg (Torr) or 14.696 psi). The atmospheric pressure decreases with increasing altitude and increases at altitudes below sea level. Weather patterns with low and high fronts decrease or increase atmospheric pressure. An aneroid barometer provides an absolute pressure measurement.

An altimeter is an absolute pressure gauge (measurement) that shows height above sea level. The air pressure to height above sea level conversion is frequently performed using a look up table. For example, 10,000 feet above sea level is 10.1 psia (69.7 kPa). Pressure altitude (*h** _{alt}*) can be calculated using the equation:

*h*_{alt}* *= (1-(*p** _{sta}*/1013.25)^0.190284)x145366.45 Eq. 1

Where:

*h** _{alt}* is the height in feet

and *p** _{sta}* is the pressure in millibars (mb) or hectopascals (hPa)

**Height of a Liquid Column**

For a standing liquid, the absolute pressure at a depth H in a liquid is defined as:

P_{abs} = P + (ρ x g x H) Eq. 2

Where:

P_{abs} is the absolute pressure at depth H in kg/m-s^{2} (or Pa)

P is the external pressure at the top of the liquid, typically atmospheric pressure for open systems

ρ is the density of the fluid (for example, pure water is 1 g/cm^{3 }at 4°C and salt water is 1.025 g/cm^{3})

g is the acceleration due to gravity (g = 9.81 m/s^{2}) (32.174 ft/s^{2}))

and H is the depth in meters or feet

**Depth in Water**

The pressure on submerged bodies increases based on the density of the liquid and depth in the liquid according to Eq. 2. Common measurements include depth in fresh or salt water. For fresh water, pressure increases 0.43 psi per foot and in salt water it is 0.44 psi per foot. A scuba diver’s submersible pressure gauge (SPG) or depth gauge makes an absolute pressure reading. A dive computer provides the time required for a safe ascent, since even a depth of 100 feet can produce a pressure of 400 kPa (3.951 atmospheres, or 58.1 psi).

**Flow in a Pipe/Tube**

Several factors determine the pressure drop that occurs in fluid flow applications including laminar versus turbulent flow, the flow velocity, kinematic viscosity and Reynolds number of the fluid, internal roughness of the inside of the pipe as well as its diameter, length and form factor. Orifice plates, venturi tubes and nozzles simplify the situation. In these cases (refer to Figure 2), the flow is related to ΔP (P_{1}-P_{2}) by the equation:

q* = *c_{d}* *π/4 D_{2}^{2} [2(P_{1}* - *P_{2}) / ρ(1 - d^{4}) ]^{1/2}* *

Where:

q is the flow in m^{3}/s

c_{D} is the discharge coefficient, the area ratio = A_{2} / A_{1}

P_{1} and P_{2} are in N/m^{2}

ρ is the fluid density in kg/m^{3}

D_{2} is the orifice, venturi or nozzle inside diameter (in m)

p1"> D_{1} is the upstream and downstream pipe diameter (in m)

and d = D_{2} / D_{1} diameter ratio

**Figure 2. **Elements of a ΔP flow measurement.

**Pressure Parameters **

As the examples demonstrate, pressure measurements use a variety of pressure standards to express the reading depending on the application. To convert between different pressure standards:

http://www.wrh.noaa.gov/slc/projects/wxcalc/formulas/pressureConversion.pdf